Also, I would contact Keithley concerning the functionality of the and A. Message 6 of The problem happened when I chose the sweep in terms of voltage rather than current. I want to edit the program and make it convenient to me, as I suppose with my limited labview knowledge it will take more time if I strat from scratch. But it seems trying to locate a resistance value for the device under test , that is unable to accomplished on a transistor. Message 2 of Message 8 of
|Date Added:||10 July 2010|
|File Size:||43.10 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Unfortunaltely keithley engineers are strongly committed not to provide the block diagram.
Message 2 of I think that the plus with the conductance measurement software from Keithley are designed for resistor-like device: Nudge I like Keithley intsruments but they are going to obsolete themselves if they don’t snap to it and write drivers that work. At this point, you may want to contact Keithley.
Keithley 2182 Meter
I noticed you selected 0 A for both the current start and stop values. Message 7 of This may be the source of the warning; the sweep is starting and stopping at 0 A. This voltage sweep function works fine over a resistor, but not over a transistor.
It is an example software written by the Keithley. Message 6 of You can contact an engineer at Keithley here. Can you tell me exactly what is “auto-test sequence running? Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type. Appreciates for any advices.
The problem happened when I chose the sweep in terms of voltage rather than current. Message keithlley of I am trying to use the “differential conductance mode” in software “KI Message 8 of As it says in the warning message, you may also want to check the connections of your transistor and verify that it is enough signal to be measured.
Keithley Nanovoltmeter – Instrument Driver – National Instruments
Thanks for your reply. Also, I would contact Keithley concerning the functionality of the and A.
Most Active Software Boards: It is possible that there is some setting that needs to be made on these particular devices to voltage sweep a resistor versus a transistor. Good luck in your applications! Please attach this VI to the reply of this post or point me in the direction to where you had found it.
For some reason, I was unable to open the data sheets and manuals on Keithley’s website. Message 3 of Since this is an executable that has been created by Keithlwy, there is no way that anyone other than Keithley can provide access to the code.
Message 10 of I don’t know if you can setup such a “transistor”measurement!?
Is this a VI that you had created? There is no manual of KI This is what I am thinking so far. Message 5 of